Electronic Components Datasheet Search
MAX11634 Datasheet(PDF) 19 Page - Maxim Integrated Products
MAXIM [Maxim Integrated Products]
MAX11634 Datasheet(HTML) 19 Page - Maxim Integrated Products
/ 24 page
12-Bit, 300ksps ADCs with Differential
Track/Hold, and Internal Reference
Internally Timed Acquisitions and
Performing Conversions in Clock Mode 00
In clock mode 00, the wake-up, acquisition, conversion,
and shutdown sequences are initiated through CNVST
and performed automatically using the internal oscilla-
tor. Results are added to the internal FIFO to be read
out later. See Figure 4 for clock mode 00 timing.
Initiate a scan by setting CNVST low for at least 40ns
before pulling it high again. The MAX11634–MAX11637
then wake up, scan all requested channels, store the
results in the FIFO, and shut down. After the scan is
complete, EOC is pulled low and the results are avail-
able in the FIFO. Wait until EOC goes low before pulling
CS low to communicate with the serial interface. EOC
stays low until CS or CNVST is pulled low again.
Do not initiate a second CNVST before EOC goes low;
otherwise, the FIFO can become corrupted.
Externally Timed Acquisitions and
Internally Timed Conversions with
Performing Conversions in Clock Mode 01
In clock mode 01, conversions are requested one at a
time using CNVST and performed automatically using
the internal oscillator. See Figure 5 for clock mode 01
Setting CNVST low begins an acquisition, wakes up the
ADC, and places it in track mode. Hold CNVST low for
at least 1.4µs to complete the acquisition. If the internal
reference needs to wake up, an additional 65µs is
required for the internal reference to power up. If a tem-
perature measurement is being requested, reference
power-up and temperature measurement are internally
timed. In this case, hold CNVST low for at least 40ns.
Set CNVST high to begin a conversion. After the con-
version is complete, the ADC shuts down and pulls
EOC low. EOC stays low until CS or CNVST is pulled
low again. Wait until EOC goes low before pulling CS or
If averaging is turned on, multiple CNVST pulses need
to be performed before a result is written to the FIFO.
Once the proper number of conversions has been per-
formed to generate an averaged FIFO result, as speci-
fied by the averaging register, the scan logic
automatically switches the analog input multiplexer to
the next requested channel. The result is available on
DOUT once EOC has been pulled low.
(UP TO 514 INTERNALLY CLOCKED ACQUISITIONS AND CONVERSIONS)
SET CNVST LOW FOR AT LEAST 40ns TO BEGIN A CONVERSION.
Figure 4. Clock Mode 00
Does ALLDATASHEET help your business so far?
[ DONATE ]
All Rights Reserved©
| English :
| Chinese :
| German :
| Japanese :
| Korean :
| Spanish :
| French :
| Italian :
| Polish :
| Vietnamese :